ELISA Kit for Osteopontin (OPN)
SPP1; BNSP; BSPI; ETA1; Secreted Phosphoprotein 1; Bone Sialoprotein I; Early T-Lymphocyte Activation 1; Nephropontin; Urinary stone protein; Uropontin
- Product No.SEA899Hu
- Organism SpeciesHomo sapiens (Human) Same name, Different species.
- Test MethodDouble-antibody Sandwich
- Assay Length3h
- Detection Range0.625-40ng/mL
- SensitivityThe minimum detectable dose of this kit is typically less than 0.237ng/mL.
- Sample Typeserum, plasma, tissue homogenates, cell lysates, cell culture supernates and other biological fluids.
- Download Instruction Manual
- UOM 48T96T 96T*5 96T*10 96T*100
For more details, please contact local distributors!US$ 468
For more details, please contact local distributors! US$ 2106
For more details, please contact local distributors! US$ 3978
For more details, please contact local distributors! US$ 32760
For more details, please contact local distributors!
- Packages (Simulation)
- Packages (Simulation)
- Results demonstration
- Typical Standard Curve
- ISO9001: 2008, ISO13485: 2003 Registered
Specificity of the ELISA Kit for Osteopontin (OPN)
This assay has high sensitivity and excellent specificity for detection of Osteopontin (OPN).
No significant cross-reactivity or interference between Osteopontin (OPN) and analogues was observed.
Recovery of the ELISA Kit for Osteopontin (OPN)
Matrices listed below were spiked with certain level of recombinant Osteopontin (OPN) and the recovery rates were calculated by comparing the measured value to the expected amount of Osteopontin (OPN) in samples.
|Matrix||Recovery range (%)||Average(%)|
Precision of the ELISA Kit for Osteopontin (OPN)
Intra-assay Precision (Precision within an assay): 3 samples with low, middle and high level Osteopontin (OPN) were tested 20 times on one plate, respectively.
Inter-assay Precision (Precision between assays): 3 samples with low, middle and high level Osteopontin (OPN) were tested on 3 different plates, 8 replicates in each plate.
CV(%) = SD/meanX100
Linearity of the ELISA Kit for Osteopontin (OPN)
The linearity of the kit was assayed by testing samples spiked with appropriate concentration of Osteopontin (OPN) and their serial dilutions. The results were demonstrated by the percentage of calculated concentration to the expected.
Stability of the ELISA Kit for Osteopontin (OPN)
The stability of kit is determined by the loss rate of activity. The loss rate of this kit is less than 5% within the expiration date under appropriate storage condition.
To minimize extra influence on the performance, operation procedures and lab conditions, especially room temperature, air humidity, incubator temperature should be strictly controlled. It is also strongly suggested that the whole assay is performed by the same operator from the beginning to the end.
Assay procedure summary of the ELISA Kit for Osteopontin (OPN)
1. Prepare all reagents, samples and standards;
2. Add 100µL standard or sample to each well. Incubate 1 hours at 37°C;
3. Aspirate and add 100µL prepared Detection Reagent A. Incubate 1 hour at 37°C;
4. Aspirate and wash 3 times;
5. Add 100µL prepared Detection Reagent B. Incubate 30 minutes at 37°C;
6. Aspirate and wash 5 times;
7. Add 90µL Substrate Solution. Incubate 10-20 minutes at 37°C;
8. Add 50µL Stop Solution. Read at 450nm immediately.
Test principle of the ELISA Kit for Osteopontin (OPN)
The test principle applied in this kit is Sandwich enzyme immunoassay. The microtiter plate provided in this kit has been pre-coated with an antibody specific to Osteopontin (OPN). Standards or samples are then added to the appropriate microtiter plate wells with a biotin-conjugated antibody specific to Osteopontin (OPN). Next, Avidin conjugated to Horseradish Peroxidase (HRP) is added to each microplate well and incubated. After TMB substrate solution is added, only those wells that contain Osteopontin (OPN), biotin-conjugated antibody and enzyme-conjugated Avidin will exhibit a change in color. The enzyme-substrate reaction is terminated by the addition of sulphuric acid solution and the color change is measured spectrophotometrically at a wavelength of 450nm ± 10nm. The concentration of Osteopontin (OPN) in the samples is then determined by comparing the O.D. of the samples to the standard curve.
Single-component Reagents of Assay Kit
Lysis Buffer Specific for ELISA / CLIA
Quality Control of ELISA / CLIA
ELISA Kit Customized Service
Disease Model Customized Service
Serums Customized Service
TGFB1 Activation Reagent
Real Time PCR Experimental Service
Fast blue Protein Stain solution
Single-component Reagents of Magnetic Luminex Assay Kit
|Catalog No.||Organism species: Homo sapiens (Human)||Applications (RESEARCH USE ONLY!)|
|EPA899Hu61||Eukaryotic Osteopontin (OPN)||Positive Control; Immunogen; SDS-PAGE; WB.|
|RPA899Hu02||Recombinant Osteopontin (OPN)||Positive Control; Immunogen; SDS-PAGE; WB.|
|EPA899Hu62||Eukaryotic Osteopontin (OPN)||Positive Control; Immunogen; SDS-PAGE; WB.|
|APA899Hu61||Active Osteopontin (OPN)||Cell culture; Activity Assays.|
|PAA899Hu06||Polyclonal Antibody to Osteopontin (OPN)||WB; IHC; ICC; IP.|
|PAA899Hu02||Polyclonal Antibody to Osteopontin (OPN)||WB; IHC; ICC; IP.|
|LAA899Hu71||Biotin-Linked Polyclonal Antibody to Osteopontin (OPN)||WB; IHC; ICC.|
|MAA899Hu22||Monoclonal Antibody to Osteopontin (OPN)||WB; IHC; ICC; IP.|
|SEA899Hu||ELISA Kit for Osteopontin (OPN)||Enzyme-linked immunosorbent assay for Antigen Detection.|
|AEA899Hu||ELISA Kit for Anti-Osteopontin Antibody (Anti-OPN)||Enzyme-linked immunosorbent assay for Antibody Detection.|
|SCA899Hu||CLIA Kit for Osteopontin (OPN)||Chemiluminescent immunoassay for Antigen Detection.|
|LMA899Hu||Magnetic Luminex Assay Kit for Osteopontin (OPN) ,etc.||Magnetic Luminex Assay for Antigen Detection.|
|KSA899Hu01||ELISA Kit DIY Materials for Osteopontin (OPN)||Main materials for "Do It (ELISA Kit) Yourself".|